HEINE® FlexTip+ F. O. Laryngoscope Blade
- Swivelling blade tip
- Clear view
- 5 years guarantee
Sets for HEINE® FlexTip+ F. O. Laryngoscope Blade
Set complete with: FlexTip+ F.O. Blades Mac 3, Mac 4
with EasyClean LED Battery Handle, zipper caseF-230.18.820
HEINE FlexTip+ Mac 3 and Mac 4
A useful Tip for a difficult intubation
115mm/137mm length, 13.2mm distal widthF-000.22.316
LED FlexTip+ Sets Mac 3, Mac 4
with Standard F.O. LED Battery HandleF-230.18.860
HEINE® FlexTip+ Fiber Optic (F.O.) Laryngoscope Blade Set
FlexTip+ F.O. Laryngoscope Set complete with: Blades Mac 3, Mac 4, spare XHL Xenon Halogen bulb case with Standard F.O. 4 NT Rechargeable Handle with Li-ion Battery and NT 4 table charger.F-230.23.865
HEINE® FlexTip+ Fiber Optic (F.O.) Laryngoscope Blade Sets
FlexTip+ F.O. Laryngoscope Set complete with: Blades Mac 3, Mac 4, with LED illumination, case, with Standard F.O. 4 LED NT Rechargeable Handle with Li-ion Battery and NT 4 table chargerF-230.94.865
FlexTip+ Sets Mac 3, Mac 4
with Standard F.O. Battery HandleF-230.10.860
Accessories for HEINE® FlexTip+ F. O. Laryngoscope Blade
EasyClean LED Laryngoscope Handle
Complete handle (without batteries)F-008.22.820
Standard F.O. 4 LED NT handle
Complete handle with rechargeable battery.F-008.22.894
F.O. 4 SLIM LED NT handleComplete handle with rechargeable battery. F-008.22.804
F.O. 4 SHORT LED NT handleComplete handle with charging module. F-008.22.424
NT4 table chargerwithout handle, with adaptor for BETA4 NT rechargeable handles, Standard F.O. 4 NT and F.O. 4 SHORT NT laryngoscope rechargeable handles. X-002.99.494
Casefor FlexTip+ F.O. Laryngoscope Set F-227, F-229 or F-230 F-227.01.000
FAQs - Frequently Asked Questions
What handles can be used with my HEINE Laryngoscope Blade?
All HEINE Laryngoscope handles can be used with the HEINE Laryngoscope Blade.
How can I clean, disinfect and sterilize my HEINE Laryngoscope Blade?
Procedure after use:
Remove all traces of contamination immediately after use to avoid incrustation.
Manual cleaning / disinfection:
Use only cleaning and disinfection products which are recommended by the manufacturer for use with fiber optics. Follow the manufacturers instructions regarding concentration and duration of use exactly. After this process, rinse the blade with de-mineralised water and ensure that no deposits remain on them. Warning: Use of the wrong cleaning and disinfection products or methods can cause irreparable damage to the laryngoscope. Ultrasonic cleaning is not permitted!
Automatic cleaning / disinfection:
The laryngoscope can be cleaned or thermally disinfected at up to 93 °C in an automatic machine. Use only cleaning products, which are recommended for use with stainless steel. The rinsing procedure must ensure that no deposits are left on the product. Use only de-mineralised water for the final rinse.
Laryngoscopes have to be sterilized as required by national regulation and internal institutional standards. We recommend the sterilization with saturated steam at least 132 °C (max. 134 °C) for 3 minutes. Cycles of up to 18 minutes or up to 60 minutes in specified cases should not damage the metal part of the blade. There may be a reduction in light transmission with frequent / repeated long-term steam sterilization, particularly if the steam quality is not ideal.
How to purchase a HEINE InstrumentFind an Authorized Distributor
HEINE Instruments are available through a worldwide network of authorized distibution partners. To locate your local authorized dealer, click above.