HEINE Modular+ Macintosh Fiber Optic Blades
Mac Modular+ F.O. Laryngoscope Blades: Both blade and light module are autoclavable and can be fitted or removed in seconds without a screwdriver. The light module does not project into the line of sight, making intubation safer and easier.
HEINE HiLite fiber bundle for improved light transmission and longer life. Modular construction – swap modules in a few seconds. Ideal Mac shape with no light pipe in the line of sight. Blade module made entirely of stainless steel. Economical modular design. Compatible with all handles to the ISO 7376 (green standard). High-quality design from chrome-plated, stainless, high-grade steel. 5 years guarantee. 4000 standard autoclave cycles of 134 °C / 4 min. with more than 1000 lux.
Sets for HEINE Modular+ Macintosh Fiber Optic Blades
Set complete with: F.O. Modular+ blade Mac 1, Mac 2, Mac 3 and Mac 4
with EasyClean LED Battery Handle, zipper caseF-228.18.820
Modular+ Sets Mac 1, Mac 2, Mac 3, Mac 4
with Standard F.O. Battery HandleF-228.10.860
HEINE® Mac Modular+ Fiber Optic (F.O.) Laryngoscope Set
Laryngoscope Set complete with: F.O. Modular+ blade Mac 1, Mac 2, Mac 3 and Mac 4, spare XHL Xenon Halogen bulb, zipper case, with Standard F.O. 4 NT Rechargeable Handle with Li-ion Battery and NT 4 table chargerF-228.23.865
LED Modular+ Sets Mac 1, Mac 2, Mac 3, Mac 4
with Standard F.O. LED Battery HandleF-228.18.860
HEINE® Mac Modular+ Fiber Optic (F.O.) Laryngoscope Sets
Laryngoscope Set complete with: F.O. Modular+ blade Mac 1, Mac 2, Mac 3 and Mac 4 with LED illumination, zipper case, with Standard F.O. 4 LED NT Rechargeable Handle with Li-ion Battery and NT 4 table charger.F-228.94.865
Accessories for HEINE Modular+ Macintosh Fiber Optic Blades
EasyClean LED Laryngoscope Handle
Complete handle (without batteries)F-008.22.820
Standard F.O. 4 LED NT handle
Complete handle with rechargeable battery.F-008.22.894
F.O. 4 SLIM LED NT handleComplete handle with rechargeable battery. F-008.22.804
F.O. 4 SHORT LED NT handleComplete handle with charging module. F-008.22.424
F.O. Angled HandleComplete handle wtih rechargeable battery. F-007.22.945
NT4 table chargerwithout handle, with adaptor for BETA4 NT rechargeable handles, Standard F.O. 4 NT and F.O. 4 SHORT NT laryngoscope rechargeable handles. X-002.99.494
HEINE Modular+ Mac 3F.O. Light module F-000.22.563
HEINE Modular+ Mac 3Blade module, metal F-000.22.533
HEINE HiLite fiber optic bundles with enhanced light transmission and long life.
Available in a full assortment of MAC and Miller sizes.
Light modules remove quickly and easily for cleaning. No screws or tools required and no parts that can be lost.
ISO 7376. Compatible with all green standard handles.
Compatible with XHL or LED handle systems.
5 year warranty. HEINE HiLite fiber optic bundle is guaranteed for 5 years or 4,000 standard autoclave cycles (134°C / 4min). Light output will not fall below 1,000 lux.
Blade frame made of 304 grade stainless steel. Strong and durable for a lifetime of maintenance free service.
Removable light modules are fully autoclaveable.
FAQs - Frequently Asked Questions
What handles can be used with my HEINE Laryngoscope Blade?
All HEINE Laryngoscope handles can be used with the HEINE Laryngoscope Blade.
How can I clean, disinfect and sterilize my HEINE Laryngoscope Blade?
Procedure after use:
Remove all traces of contamination immediately after use to avoid incrustation.
Manual cleaning / disinfection:
Use only cleaning and disinfection products which are recommended by the manufacturer for use with fiber optics. Follow the manufacturers instructions regarding concentration and duration of use exactly. After this process, rinse the blade with de-mineralised water and ensure that no deposits remain on them. Warning: Use of the wrong cleaning and disinfection products or methods can cause irreparable damage to the laryngoscope. Ultrasonic cleaning is not permitted!
Automatic cleaning / disinfection:
The laryngoscope can be cleaned or thermally disinfected at up to 93 °C in an automatic machine. Use only cleaning products, which are recommended for use with stainless steel. The rinsing procedure must ensure that no deposits are left on the product. Use only de-mineralised water for the final rinse.
Laryngoscopes have to be sterilized as required by national regulation and internal institutional standards. We recommend the sterilization with saturated steam at least 132 °C (max. 134 °C) for 3 minutes. Cycles of up to 18 minutes or up to 60 minutes in specified cases should not damage the metal part of the blade. There may be a reduction in light transmission with frequent / repeated long-term steam sterilization, particularly if the steam quality is not ideal.
How to purchase a HEINE InstrumentFind an Authorized Distributor
HEINE Instruments are available through a worldwide network of authorized distibution partners. To locate your local authorized dealer, click above.