HEINE ML4 LED HeadLight with HR / HRP Optics
Perfect fit. Perfect view. During long procedures.
Topseller in all surgical specialties: The HEINE ML4 LED HeadLight is specially designed for healthcare professionals who need a perfect view during long surgical operations or treatments.
- Perfect fit
- Shadow free
- Adjustable spot size
- LED in HEINE Quality – or LED HQ.
- HEINE Binocular Loupes and Headlights ordering information
Topseller in all surgical specialties: The HEINE ML4 LED HeadLight is specially designed for healthcare professionals who need a perfect view during long surgical operations or treatments. The headband of the ML4 LED HeadLight offers multiple adjustments points and soft padding ensuring highest comfort and a firm fit.
Catalogue Number: J-008.31.445
ML4 LED HeadLight on Professional L headband and Power Sources , HR Binocular Loupe 2.5x/340mm, Wall Transformer EN 50 UNPLUGGED, 2 x mPack UNPLUGGED, S-GUARD, Cleaning Fluid
- HEINE Headlights HygienicReprocessing_EN_GB.pdf
- Brochure Binocular Loupes and Headlights
- HEINE Instruction for Use_ML4LED_HeadLight_MED113192
- User report HEINE ML4 LED Dr.Berchtold
- Instruction for Use HR/ HRP/ HR-C Binocular Loupes
- User report HEINE ML4 Dr. José Andrés Vela
- User Report HEINE ML4 Dr.Tschandl
FAQ - Frequently Asked Questions
How can I clean and disinfect the HEINE Binocular Loupes?
Manual Cleaning: Before cleaning, remove the lens protectors. Use a microfibre cleaning cloth when removing fingerprints from the optics. Use a damp cloth when cleaning the housing and the optics (water, cleaning solutions with a neutral or mildly alkaline pH). The loupes are water resistant and can be washed under running water. If residues/dirt must be removed from the optics and/or the housing, use a soft brush.
Manual Desinfection: Disinfection by means of spraying or wiping is permitted. Apply cleaning solutions that have been approved for medical devices made from plastic and glass. Apply isopropyl alcohol (only up to a maximum of 70% concentration). Warning: Immersion disinfection, machine cleaning/disinfection, ultrasound cleaning and sterilisation are not permitted.
How can I clean and disinfect my HEINE ML4 LED HeadLight?
Clean and disinfect the head lights, the accessories and the headbands manually (wipe-cleaning and wipe-disinfection). Clean the leather cushion of the headbands with a wet wipe. Conditioning residues on optics may be removed using a low-bacterial, dry, microfiber cloth. Recommended agents: Cleaning agent: Neodisher® MediClean Disinfectant agent: quaternary ammonia compounds (e.g. Microbac® tissues) Polarisation filter P2 may be reprocessed up to 50 times
Can I connect a camera to my ML4 LED HeadLight?
Yes, you can connect the HEINE DV1 to your ML4 LED HeadLight. For further information please contact our HEINE distributor or visit our online distributor locator.
How do I change the light source in my HEINE ML4 LED HeadLight?
The HEINE LEDs feature an extremely long durability, sustaining their illumination quality throughout the whole product life time. Therefore an exchange of the LED is not necessary.
How can I control and change the illumination spot of my ML4 LED HeadLight?
The HEINE ML4 LED HeadLight has an adjustable illumination spot size with a spot size range of 30 mm to 80 mm (at a working distance of 420 mm). The spot size can be controlled via the aperture lever on the headlight.
Find a distributor
Are you interested in one of our products?
Find a distributor near you quickly and easily.
As a global leader in the manufacture of primary diagnostic instruments with over 500 employees, HEINE Optotechnik has been a 100 % family owned and managed company for more than 70 years. We continue to develop and manufacture HEINE instruments at our facilities in Germany, where we combine decades-long experience and craftsmanship with the most modern manufacturing technologies.Quality Made in Germany